Imatest SFRreg performs highly automated measurements of sharpness (expressed as Spatial Frequency Response (SFR), also known as Modulation Transfer Function (MTF)) and Lateral Chromatic Aberration from images that contain registration mark patterns (circles that contain two light and two dark quadrants).
This document illustrates SFRreg results. Part 1 introduced SFRreg and explained how to obtain and photograph the chart. Part 2 showed how to run SFRreg inside Rescharts and how to save settings for automated runs.
When calculations are complete, results are displayed in the Rescharts window, which allows a number of displays to be selected. The following table shows where specific results are displayed.
|MTF (sharpness) for individual regions||1. Edge and MTF|
|MTF (sharpness) for entire image||4. Multi-ROI summary
11. 3D plot
12 Radial (Lens-style) MTF plot
|Lateral Chromatic Aberration||2. Chromatic Aberration|
|Original image showing region selection||8. Image & geometry|
|EXIF data||7. Summary & EXIF data|
|Acutance/SQF (Subjective Quality Factor)||3. Acutance / SQF|
|Edge roughness||13. Edge roughness|
|Chromatic Aberration (radial)||14. Radial (Chr Aber, etc.)|
|Tonal response, Gamma/White Balance||5. Tonal response…||A|
|Color analysis||9. a*b* Color error
10. Split colors: Reference/input
|Noise analysis||15. Noise||A|
Note A: Available in Imatest 5.0+ when the 3 row x 5 column chart is detected.
Multi-ROI summary display
The multi-ROI (multiple Region of Interest) summary shown in the Rescharts window (above) contains a detailed summary of SFRreg results. (3D plots also contain an excellent summary.) The upper left contains the image in muted gray tones, with the selected regions surrounded by red rectangles and displayed with full contrast. Up to four results boxes are displayed next to each region. The results are selected the the Display options area on the right of the window, below the Display selection.
The Results selection (right) lets you choose which results to display. N is region number. Ctr-corner distance % is the approximate location of the region. CA is Chromatic Aberration in area, as percentage of the center-to-corner distance (a perceptual measurement). A legend below the image shows which results are displayed.
The View selection (far right) lets you select how many results boxes to display, which can be helpful when many regions overlap. From top to bottom the number of boxes is 4, 3, 2, 2, and 1, respectively.
The approximate location of maximum MTF, based on a second order fit to MTF results, is indicated by a red ‘O’ at the intersection of vertical and horizontal dotted lines (– – – – O – – – –).
Distortion statistics, which are displayed for SFRplus and eSFR ISO, are not available for SFRreg.
MTF asymmetry is calculated in the x (horizontal) and y (vertical) directions. MTF50 as functions of x and y are fitted to second-order (parabolic) curves, which are used to calculate expected MTF50 values at the Top, Bottom, Left and Right (MTF50T, etc.) of the image.
MTF asymmetry (x) = (MTF50R-MTF50L) / (MTF50R+MTF50L)
MTF asymmetry (y) = (MTF50T-MTF50B) / (MTF50T+MTF50B)
Weighted summary results are displayed on the upper-right. The default weights are 1.0 for ROIs in the central region (inside the inner dotted circle), 0.75 for the middle region (between the two dotted circles), and 0.5 for the outer region (outside the outer dotted circle). These weights can be changed in the SFRreg Settings window. The results are independent of the number of ROIs in each region.
Plot settings area
A small number of options are available in the Plot settings area, on the lower right, below the Display box (fewer than for other Rescharts modules).
All displays contain the dropdown menu for selecting the primary channel to analyze. If R, G, B, or Y (Luminance) is selected, all channels are analyzed, but the selected channel is emphasized. There is also an option to analyze any of the channels alone— useful where one of the secondary channels is dark and may cause a run to crash. Luminance (Y) is shown above. If it is changed, results are recalculated.
Many displays allow you to select the ROI for viewing results.
The Edge and MTF display has a dropdown window for selecting the maximum MTF display frequency: 2x Nyquist (the default), Nyquist, 0.5x Nyquist, and 0.2x Nyquist.
Edge and MTF display
This display is identical to the SFR Edge and MTF display. The edge (or line spread function) is plotted on the top and the MTF is plotted on the bottom. The edge may be displayed linearized and normalized (the default; shown), unlinearized (pixel level) and normalized, or linearized and unnormalized (good for checking for saturation, especially in images with poor white balance). Edge display is selected by pressing .
There are a number of readouts, including 10-90% rise distance, MTF50, MTF50P (the spatial frequency where MTF is 50% of the peak value, differing from MTF50 only for oversharpened pulses), the secondary readouts (MTF @ 0.125 and 0.25 C/P in this case), and the MTF at the Nyquist frequency (0.5 cycles/pixel). The diffraction-limited MTF curve (not shown above) is displayed as a pale brown dotted line when pixel pitch is entered.
An important (and optional) readout in the upper plot is
- The Chart contrast (entered in the Display options section of the SFRreg setup and SFRreg More settings windows),
- The Contrast factor: the ratio between the chart contrast derived from the pixel ratio and the input value of gamma (0.5 in the above display),
- Gamma (from chart): a measurement of gamma derived from the chart pixel levels and the Chart contrast (an input value, as described above). Gamma has a relatively small effect on the MTF measurement, especially for moderate to low contrast targets (4:1 or under).
Lateral Chromatic Aberration (LCA), also known as “color fringing,”
is most visible on tangential boundaries near the edges of the image. Much of the plot is grayed out if the selected region (ROI) is too close to the center (less than 30% of the distance to the corner) to accurately measure CA.
The area between the highest and lowest of the edge curves (shown for the R, G, B, and Y (luminance) channels) is a perceptual measurement of LCA. It has units of pixels because the curves are normalized to an amplitude of 1 and the x-direction (normal to the edge) is in units of pixels. It is displayed as a magenta curve.
Perceptual LCA is also expressed as percentage of the distance from center to corner, which tends to be more reflective of system performance: less sensitive to location and pixel count than the pixel measurement. Values under 0.04% of the distance from the center are insignificant; LCA over 0.15%
can be quite visible and serious.
The CA (crossing) numbers are highly correlated to the optical LCA
The LCA crossing distances (R-G and B-G) are also given in pixels and units of % (center-to-corner). They correlate well with the optical LCA— a property of the optics, affected very little by demosaicing. LCA can be corrected most effectively before demosaicing. Results are explained in Chromatic Aberration … plot.
|Acutance and SQF are perceptual measurements of the sharpness of a display (monitor image or print). MTF, by comparison is device sharpness (not perceptual sharpness). Acutance and SQF includes the effects of the human visual system’s Contrast Sensitivity Function (CSF), display size, and viewing distance, which can be set with the Acutance/SQF Options button in More settings. Acutance/SQF are available when Speedup is unchecked in the input dialog box. See Introduction to Acutance and SQF for more detail.|
Histograms and noise analysis
Histograms and noise stats
This display, available when Speedup is unchecked, contains histograms of pixel levels for individual ROIs (original on top and linearized using input gamma on bottom). Since this slows calculations considerably, we only recommend unchecking Speedup when histograms are needed (which will be infrequently).
The black (background) histogram contains pixel levels for the entire ROI. The red histogram is for the right (dark) region, away from the transition, used in the noise statistics calculation. The cyan histogram is for the left (light) region. Sharpening may cause extra bumps to appear in the black histogram. A detailed level and noise analysis is displayed below the image.
|Dark, light levels||Original pixel levels normalized to 1 and linearized levels, also normalized to 1|
|Estimated chart contrast (for gamma = …)||The correct chart contrast must be entered in SFRplus parameters & setup window.|
|Noise (dark, light, mean; norml pixels %)||RMS noise expressed in pixels, normalized to 100% (for 255 in 8-bit files)|
|S/N (…; norml pixels)||Signal/Noise, where signal is mean pixel level of ROI at a distance from the transition.|
|Noise (…; linearized %)||Noise, linearized using gamma (input); normalized to 100%|
|S/N (…; linearized)||Signal/Noise, linearized|
Noise calculations are made in portions of the ROIs (Regions of Interest) away from the transitions. They are facilitated by selecting wide ROIs. The noise spectrum on the lower right contains qualitative information about noise visibility and software noise reduction, which generally reduces high frequency noise below 0.5, typical of demosaicing alone. The region on the right is shown in red; left is shown in cyan. See also Noise.
Note: This display has much less geometric detail than for SFRplus and eSFR ISO, which have highly structured charts that allow distortion, Field of View, and geometric shifts to be calculated and displayed.
It shows the selected regions for MTF/noise analysis (violet rectangles). Display options include
|Original Image||Extreme Lighten (HSL)|
|Rec channel||Lighten (HSV)|
|Green channel||Lighten 1 bit|
|Blue channel||Lighten 2 bits|
|Color: Boost Sat. 4X||Original w/5×5 grid|
|Color: Boost Sat, 10X||Original w/5×7 grid|
|Lighten (HSL)||Original w/11×15 grid|
A very limited number of geometrical results are shown beneath the image, including
- WxH of image in pixels; m x n squares found; ROI size (referencing input setting), number of ROIs for MTF, etc. (not for profiles).
A particularly interesting display is available when you click Crop to ROI (or Crop to 1/2 ROI). The ROI is displayed large enough so its blur is clearly visible and can be compared with Edge and MTF plots and summary results.
Summary & EXIF data
|This plot contains summary results for individual ROIs and EXIF data (metadata that describes camera and lens settings). When first installed, Imatest reads EXIF data from JPEG files only. Enhanced EXIF support requires a special download of ExifTool, described here.EXIF data may be viewed at any time from any display by clicking File, View All EXIF Data.||
Summary & EXIF data
|Displays results as three-dimensional pseudocolor images. The parameter to be displayed is selected by Plot options, shown on the right. (Only MTF50, automatically scaled, can be displayed in Imatest Studio) The display can be rotated for enhanced visualization. Region selection should include at least 13 regions.A 5x5 ROI grid (23 regions) was chosen for the analysis on the right. The largest selection that avoids the low contrast edges, which tend to have lower MTFs in typical consumer/DSLR cameras with nonlinear signal processing, is 13. All sqs, inner & bdry except low contrast. Single-contrast charts (without the low contrast regions) work well for 3D plots (for example, in equipment reviews). The rather strangely shaped surface (center has lower MTF than part way region) may be the result of autofocus issues and/or curvature of field.Several Display options are available. The Pseudocolor shaded plot with image and text shown on the right is the most generally useful. Other options include no shading, colored contour lines-only, and 3D plots only (displayed larger, but without the summary image and text shown below the 3D plot).
Display area. Details on right
The Z-axis of the image (which contains the results) can be inverted to reveal detail obscured in a normal display. The background color can be varied from white to black using the Backgnd slider. The default is light gray (0.9).
Manual or automatic scaling may be selected in Master using the button. Automatic is the default. When is pressed, the Manual scaling window, shown on the right, lets you set the scaling for the parameter selected in the Plot menu (MTF50P in this case). Minimum and Maximum only apply for manual scaling, i.e., when Auto scaling is unchecked. 6, 11, 16, 21, or 26 contour lines may be selected.
The button sets a top (pseudo-2D) view. It toggles between and . You can rotate the image starting from either setting.
Several 3D plots can be displayed in SFRplus auto by making appropriate selections in the SFRplus settings window. button in the
MTF50 3D plot
Because this plot works best with large numbers of edge radii, which may be difficult with registration mark targets, we will keep this description brief. For more detail, see the SFRplus results page.
This plot produces results similar to the MTF plots in the Canon (explanation), Nikon (explanation), and Zeiss websites. Up to three plot parameters may be selected with the Secondary readout. Typical choices are MTF at 10, 20, and 40 lp/mm (used by Zeiss) or 10 and 30 lp/mm (used by Canon and Nikon). Pixel spacing (um/pixel, etc.) must entered to obtain lp/mm spatial frequencies.
Differences between the Imatest Lens-style MTF plot and manufacturer’s website MTF plots are discussed on the SFRplus results page.
Lens-style MTF plot
Edge roughness plot
Analyzes edge roughness, which has two components:
- Random (aperiodic), caused by noise.
- Periodic (also called “jaggies” for jagged edges), caused by aliasing (signal response above the Nyquist frequency). Affected by the demosaicing algorithm.
The periodic noise has peak spectral response at spatial frequencies corresponding to one or two pixels perpendicular to the edge, where one pixel corresponds to the green channel and two pixels corresponds to the red and blue channels of Bayer color filter arrays. In the Roughness Spectrum (lower) plot these frequencies are 1 and 0.5, respectively. The response peaks at these frequencies are fairly typical.
The periodic noise is calculated by transforming the frequency components at f = 0.5 through 2.5 in steps of 0.5 ( i.e., the aliasing fundamentals and the first few harmonics) back into spatial domain. The aperiodic noise is calculated by subtracting the periodic noise from the total noise.
Three RMS (s = standard deviation) roughness values are reported: Total, aliasing (jaggies), and Total-aliasing (aperiodic).
The interpretation of edge roughness results is somewhat challenging because it’s a new measurement and there isn’t a lot of data for comparison. While the pattern on the right is fairly typical (and expected), we’ve seen many spectra that don’t resemble it at all. In general noise, peaks are weak (or nonexistent) for blurred edges. Although we hope this calculation will enable us to distinguish excellent and mediocre demosaicing algorithms, we’re not there yet.
Imatest can estimate an imaging system’s loss of resolution (PSF) using a pair of collocated, near-sagittal and tangential slanted edges on SFRplus, eSFR ISO and SFRreg targets. Easily analyze and improve sharpness, chromatic aberration, and distortion measurements.
When an SFRreg Center chart is detected in the image in Imatest 5.0+, four additional plots are available: Tonal response, a*b* color error, Split colors, Density response, and Noise.
An image of a lab setup consisting of an SFRplus Center chart and four additional SFRreg charts is shown on the right. The distortion is slightly larger than the Center chart was designed for, but not excessive. Everything works perfectly.
SFRreg image (click to view downloadable full-sized image)
This display is derived from the 20-patch ISO 14524 stepchart pattern, located in the center of the SFRreg Center chart. It resembles the third figure in Stepchart. The upper plot shows the tonal response (also called OECF for Opto-Electronic Conversion Function) for all colors. The lower plot shows instantaneous gamma— the slope (derivative) of tonal response. (White balance plots are also available.) The value of gamma may differ slightly from the values in the Edge response and MTF display because it’s calculated differently– based on the average slope of the light to middle tone squares of the stepchart.
The levels of the light and dark portions of the patches used to calculate MTF are shown as pale horizontal lines and as points on the extreme left and right of the upper plot. You can use these lines to see if the patches are within the (relatively) linear region of tonal response (they are in the example on the right). A warning message (Warning: MTF patches may saturate ) is displayed if any of the patches are close to saturation (log(pixel level/255) > -0.01; roughly, pixel level > 249 of 255).
SFRreg can display several types of noise, measured from the grayscale patches surrounding the center of the chart, including all types of noise calculated by Multicharts, Multitest, and eSFR ISO (standard pixel noise, chroma noise, scene-referenced noise, sensor (raw) noise, and ISO-15739 visual noise). SNR (Signal-to-Noise Ratio) can also be displayed as a ratio or in dB.
Instructions for selecting the noise display and x-axis and descriptions of the noise displays are on Multicharts/Multitest and eSFR ISO Noise.
The figure on the right— scene-referenced SNR, expressed as a simple ratio— can be used to calculate dynamic range (though it's more effective with transmissive (high tonal range) targets).
Noise plot (for scene-referenced noise,
Geometry— Additional results
When the Center chart is present, pixel shift, rotation, pixel aspect ratios, and convergence angles (related to keystone distortion) are displayed. Distortion and Field of View are not (yet) displayed.
- SFRreg analyzes images of registration mark patterns, which may be obtained from the Optikos Meridian compact projection system (where the images may be focused at infinity) or from an array or printed patterns, which can work with extreme wide angle fisheye lenses. There should be few or no interfering patterns (resembling registration marks) outside the registration mark charts.
- Lighting should be even and glare-free. Lighting and alignment recommendations are given in The Imatest test lab.
- The first time SFRreg is run, it should be run through Rescharts. This allows
- parameters to be adjusted and saved for later use in the automatic version of SFRreg, which is opened by pressing Fixed modules (dropdown menu), SFRreg Auto in the Imatest main window.
- results (listed above) to be examined interactively in the Rescharts window.
Result file names— The roots of the file names are the same as the image file name. The channel (Y, R, G, or B) is included in the file name. If a Region of Interest has been selected from a complete digital camera image, information about the location of the ROI is included in the file name following the channel. For example, if the center of the ROI is above-right of the image, 20% of the distance from the center to the corner, the characters AR20 are included in the file name.
(default location: subfolder Results)
|Excel .CSV (ASCII text files that can be opened in Excel)|
|SFR_cypx.csv||(Database file for appending results: name does not change). Displays 10-90% rise in pixels and MTF in cycles/pixel (C/P).|
|SFR_lwph.csv||(Database file for appending results: name does not change). Displays 10-90% rise in number/Picture Height (/PH) and MTF in Line Widths per Picture Height (LW/PH).|
|filename_YA17_MTF.csv orfilename_nn_MTF.csv||Excel .CSV file of MTF results for this region (designated by location (YA17) or sequence (nn = 01,...). All channels (R, G, B, and Y (luminance) ) are displayed.|
|filename_Y_multi.csv||Excel .CSV file of summary results for a multiple ROI run.|
|filename_Y_sfrbatch.csv||Excel .CSF file combining the results of batch runs (one or more files) with multiple ROIs. Only for automatic SFRplus (not Rescharts). Used as input to Batchview.|
Imatest SFR creates or updates output files for use with Microsoft Excel. The files are in CSV (Comma-Separated Variable) format, and are written to the Results subfolder by default. .CSV files are ASCII text files that look pretty ugly when viewed in a text editor:
|File ,Date/time ,PH,Ch,H/V,10-90U,10-90C,Over-,Over-,MTF50U,MTF50C,MTF,Camera,Lens,FL,f-stop,Loc,Misc.,,,,,/PH,/PH,shoot%,sharp%,LW/PH,LW/PH,Nyq,,,(mm),,,settingscanon_eos10d_sfr.jpg,2004-03-19 22:21:34, 2048,Y,H, 1422, 1447, 19.5, -0.7, 1334, 1340,0.154,,,,,canon_g5_sfr.jpg,2004-03-19 22:24:30, 1955,Y,H, 1973, 1301, 48.0, 21.3, 1488, 1359,0.268,Canon G5,—,14,5.6,ctr,sigma_sd9_sfr.jpg,2004-03-19 22:27:55, 1504,Y,H, 1432, 1676, 2.4, -7.7, 1479, 1479,0.494,,,,,sigma_sd10_sfr.jpg,2004-03-19 22:28:32, 1504,Y,H, 1563, 1628, 11.9, -2.0, 1586, 1587,0.554,,,,,|
But they look fine when opened in Excel.
.CSV files can be edited with standard text editors, but it makes more sense to edit them in Excel, where columns as well as rows can be selected, moved, and/or deleted. Some fields are truncated in the above display, and Date/time is displayed as a sequence of pound signs (#####...).
The format can be changed by dragging the boundaries between cells on the header row (A, B, C, ...) and by selecting the first two rows and setting the text to Bold. This makes the output look better. The modified file can be saved with formatting as an Excel Worksheet (XLS) file. This, of course, is just the beginning.
It's easy to customize the Excel spreadsheet to your liking. For example, suppose you want to make a concise chart. You can delete Date/time (Row B; useful when you're testing but not so interesting later) and Channel (all Y = luminance). You can add a blank line under the title, then you can select the data (rows A4 through J7 in the image below) and sort on any value you choose. Corrected MTF50 (column I) has been sorted in descending order. Modified worksheets should be saved in XLS format, which maintains formatting.
There are no limits. With moderate skill you can plot columns of results. I've said enough. ( I'm not an Excel expert! )
An optional .CSV (comma-separated variable) output file contains results for MTF and other data. Its name is [root name]_[channel location]_MTF.csv, where channel is (R, G, B, or Y) and the location BL75 means below-left, 75% of the distance to the corner (from the center). An example is Canon_17-40_24_f4_C1_1408_YBL75_MTF.csv. Excerpts are shown below, opened in Excel.
A portion of the summary CSV file, opened in Excel
The format is as follows:
|Line 1||Imatest, release (1.n.x), version (Light, Pro, Eval), module (SFR, SFR multi-ROI, Colorcheck, Stepchart, etc.).|
|File||File name (title).|
|Run date||mm/dd/yyyy hh:mm of run.|
|Tables||Separated by blank lines if more than one. Two tables are produced.|
|The first table contains MTF. The columns are Spatial frequency in Cy/mm, LW/PH, MTF (selected channel), MTF (Red), MTF (Green), MTF (Blue), MTF (Luminance = Y). (...) represent rows omitted for brevity.|
|The second table contains the edge. Columns are x (location in pixels), Red edge, Green edge, Blue edge, Luminance (Y) edge, and Chromatic Aberration (the difference between the maximum and minimum).|
|Additional data||The first entry is the name of the data; the second (and additional) entries contain the value. Names are generally self-explanatory (similar to the figures).|
|EXIF data||Displayed if available. EXIF data is image file metadata that contains important camera, lens, and exposure settings. By default, Imatest uses a small program, jhead.exe, which works only with JPEG files, to read EXIF data. To read detailed EXIF data from all image file formats, we recommend downloading, installing, and selecting Phil Harvey's ExifTool, as described here.|
This format is similar for all modules. Data is largely self-explanatory. Enhancements to .CSV files will be listed in the Change Log.
The optional XML output file contains results similar to the .CSV file. Its contents are largely self-explanatory. It is stored in [root name].xml. XML output will be used for extensions to Imatest, such as databases, to be written by Imatest and third parties. Contact us if you have questions or suggestions.
An optional .CSV file is also produced for multiple ROI runs. Its name is [root name]_multi.csv.